The microprocessor based, 1MHZ dual frequency unit withLCD Display & a Soft touch membrane keyboard enables the physician to setthe parameters easily.
Output Frequency : Two, 1MHz and 3MHz
Output Watts : Continuous Mode : 15 watts, PulsedMode : 21 watts
Intensity : 0 to 3.5 watts / cm2 maximum. USoutput can be adjusted by intensity switch.
Treatment Heads : Two. (One Small & One Largeof 5cm sq.)
Pulse Mode Ratios : 1:1 , 1:2 , 1:4 , 1:7 , 1:10 ,1:16
Effective Area of Radiation : 5 square centimeter(Large Head).
Treatment Timer : 00 to 60 minutes with Buzzer atthe end of Treatment Time.
Display : The LCD displays Parameter chosen,Treatment Time and Output Power.
Programs : 45 pre-fixed programmed
Power : 230V / AC / 50Hz. 0.6Amps.
Thank You for your valuable time. We have received your details and will get back to you shortly.